Evaluating the use of fluorescent imaging for the quantification of dental fluorosis
- Michael G McGrady†1Email author,
- Roger P Ellwood†2,
- Andrew Taylor2,
- Anne Maguire3,
- Michaela Goodwin†1,
- Nicola Boothman†1 and
- Iain A Pretty†2
© McGrady et al.; licensee BioMed Central Ltd. 2012
Received: 30 May 2012
Accepted: 25 October 2012
Published: 1 November 2012
The quantification of fluorosis using fluorescence imaging (QLF) hardware and stain analysis software has been demonstrated in selected populations with good correlation between fluorescent image metrics and TF Index scores from photographs. The aim of this study was to evaluate the ability of QLF to quantify fluorosis in a population of subjects (aged 11–13) participating in an epidemiological caries and fluorosis survey in fluoridated and non-fluoridated communities in Northern England.
Fluorescent images of the maxillary incisors were captured together with standardized photographs were scored blind for fluorosis using the TF Index. Subjects were excluded from the analysis if there were restorations or caries on the maxillary central incisors.
Data were available for 1774 subjects (n=905 Newcastle, n=869 Manchester). The data from the fluorescence method demonstrated a significant correlation with TF Index scores from photographs (Kendall’s tau = 0.332 p<0.0001). However, a number of additional confounding factors such as the presence of extrinsic stain or increased enamel translucency on some subjects without fluorosis or at low levels of fluorosis severity had an adverse impact on tooth fluorescence and hence the outcome variable. This in conjunction with an uneven distribution of subjects across the range of fluorosis presentations may have resulted in the lower than anticipated correlations between the fluorescent imaging metrics and the photographic fluorosis scores. Nevertheless, the fluorescence imaging technique was able to discriminate between a fluoridated and non-fluoridated population (p<0.001).
Despite confounding factors the fluorescence imaging system may provide a useful objective, blinded system for the assessment of enamel fluorosis when used adjunctively with photographic scoring.
The latter half of the 20th Century demonstrated a decline in the prevalence of dental caries through the use of optimally fluoridated community water supplies and fluoridated oral care products. However, this reduction in caries has also been associated with concerns regarding increased prevalence of dental fluorosis in both fluoridated and non-fluoridated communities[1–4].
In the UK, a systematic review commissioned by the government known as the York Report set out to review the safety and efficacy of water fluoridation. The report stated the occurrence of fluorosis at water fluoride levels of 1ppm was found to be high (predicted 48%, 95% CI 40 to 57). Of this fluorosis, the proportion considered to be aesthetically objectionable was lower (predicted 12.5%, 95% CI 7.0 to 21.5). A study conducted in Newcastle upon Tyne (fluoridated) and Northumberland (non-fluoridated) found increased prevalence of fluorosis in the fluoridated area compared to the non-fluoridated area with similar figures for overall fluorosis prevalence quoted in the York Report but the prevalence of aesthetically objectionable fluorosis was lower at 3.4%. The authors suggested reasons for similarities and differences in prevalence data from other studies[7, 8].
There are several possible explanations for the perceived increase in fluorosis prevalence. There could be a true increase in prevalence reflecting an increase in fluoride exposure from various sources of fluoride and an associated increased risk of fluorosis. However, there are other plausible explanations that could explain the increase in prevalence. Traditionally, fluorosis has been assessed by the use of clinical indices such as Dean’s Index and the Thylstrup & Fejerskov (TF) Index. The employment of clinical indices relies upon subjective assessment and interpretation of predetermined criteria, which may impart bias. In light of this and despite a wealth of historical data there have been criticisms of the use of clinical indices in the York Report and elsewhere in the literature[12, 13].
The choice of index may influence the investigation of fluorosis prevalence. Large volumes of data were collected through the work of H Trendley Dean utilizing an index that bore his name. This work subsequently led to the implementation of water fluoridation schemes[10, 14–16]. Despite criticism of Dean’s Index[13, 17] it remains a popular index particularly in the United States. A major difference between Dean’s Index and the TF Index is Dean’s Index assesses teeth wetted by saliva and TF Index requires the drying of teeth prior to assessment. The latter technique highlights the presence of more mild presentations of fluorosis which in itself may result in an apparent increase in fluorosis prevalence and difficulties particularly when comparisons are made to historical data using alternate indices.
An additional issue with clinical indices is the possibility of examiner bias. This may manifest through lack of blinding during assessment or variability in inter and intra-examiner agreement. There is also a phenomenon of personal thresholding particularly at low levels of fluorosis severity with differences in the application of diagnostic criteria[18, 19]. Attempts have been made to address some of the issues associated with the use of clinical indices. The remote scoring of standardized clinical photographs addresses issues pertaining to examiner blinding and facilitates the longitudinal assessment of fluorosis through the archiving of materials and repeatability of image capture[20, 21]. However, as this technique still fundamentally relies upon an examiner employing a subjective index, all of the confounding issues of a clinical index cannot be overcome. Consensus scoring of remote images may address some issues relating to personal thresholding. A further consideration of the remote scoring technique is the viewing medium for image scoring. Magnification of images may increase the detection of milder forms of fluorosis and hence affect prevalence data relative to historical data and potential prospective data if viewing conditions are not carefully controlled.
The York Report and a report from the Medical Research Council that followed both stated the evidence base of studies on water fluoridation required improvement and were critical of the use of such subjective indices for the assessment of fluorosis. Future work should consider more reliable and objective means of quantifying fluorosis severity and for longitudinal monitoring.
Recent years have seen an emphasis on the detection and quantification of dental caries utilizing emerging technologies and diagnostic sciences. The development of caries detection systems with improved sensitivity and specificity over traditional visual and tactile techniques has invigorated the field of cariology enabling more preventative interventions to be used more successfully in preventing caries and the remineralization of early carious lesions. Unfortunately, the advances within cariology have not been reflected in the study of fluorosis where clinical indices still remain the gold standard. However, consideration has been made in the literature to the application of optical techniques employed in caries detection for assessment of fluorosis. One such technique is quantitative light induced fluorescence (QLF). QLF has been investigated as a means of detecting and quantifying early enamel carious lesions[24, 25] and has since be explored as a tool for quantifying dental plaque, tooth surface loss (erosion), extrinsic stain and for the quantification of fluorosis[26–29].
Early work on the use of QLF in fluorosis quantification was encouraging. A novel software analysis technique was designed to overcome the difference in presentation of caries (discreet lesions) and fluorosis (diffuse lesions) and the resultant differences in fluorescence signal when using fluorescent imaging. On a selected population with milder forms of fluorosis, QLF achieved very good intra class correlation coefficients (ICC) when compared to the TF Index (Kendall’s Tau = 0.869). However, there were a number of confounding factors. There is an inherent difficulty in determining the potential of QLF as a means of quantifying fluorosis as there is no current acceptable gold standard with which to compare the output metrics of a fluorescent imaging system. The ordinal data derived from a subjective clinical index cannot be easily compared to the continuous data generated from QLF. Hence the analysis could only determine the association between the two techniques, not true agreement. Furthermore, as QLF relies upon the detection of changes in fluorescence between “sound” and “unsound” enamel, any artefact-inducing scattering of the reflected light from the tooth surface could result in a change in fluorescence and aberrant readings for fluorosis quantification. Such artefacts include presence of caries, extrinsic stain, restorations and non-fluorotic opacities. Nevertheless, QLF demonstrated in a small, selected population with a relatively limited range of fluorosis presentations the potential as a means of delivering a system for the reliable, objective quantification of enamel fluorosis.
Subsequent work aimed not only to refine the QLF system in fluorosis quantification by investigating alternate analysis techniques but also determining if QLF could discriminate between a wider range of presentations of fluorosis severity in larger populations with varying exposures to fluoride. The outcome of this work determined the use of a convex hull software algorithm was a more reliable means of quantifying fluorosis and that QLF could discriminate between populations with differing fluoride exposure and fluorosis severity. However, the confounding factors remained unresolved. Despite these limitations QLF still demonstrated potential as a means of objective, blinded quantification and a means of providing a system for longitudinal monitoring.
The aim of this study was to evaluate the use of fluorescent imaging for the quantification of dental fluorosis in an epidemiological survey and to determine the level of association with remote photographic scoring using a standard clinical index.
Subjects were selected for this study had participated in an epidemiological survey looking at caries and fluorosis prevalence and severity in two areas in Northern England, Newcastle upon Tyne which is has community water supplies fluoridated at an adjusted level of 1 mgF/L and Greater Manchester which receives non-fluoridated water supplies. The protocol for the study received ethical approval from the University of Manchester Committee on Ethics on Research on Human Beings (ref: 07952).
Screening and selection of participants
Subjects were healthy males and females aged 11–13 years old who were lifetime residents of their locality. Written consent was obtained from the subjects after the parents or carers had been given two opportunities to object to their child’s participation via a postal return of pre-prepared forms sent out prior to study recruitment.
Fluorescence image capture
A convex hull analysis software package originally designed to quantify stain on teeth was utilized. The software was designed to detect diffuse areas on the tooth surface using an algorithm based on a convex hull to detect and quantify the diffuse areas of hypomineralization associated with fluorosis. The application of this methodology has been described in the literature.
Data management and analysis
The data for the photographic TF index scores from the epidemiological survey were entered into the Statistical Package for Social Sciences (SPSS 16.0) along with the metrics from the analysis of the fluorescent images using convex hull software. For each subject, the higher of the two scores on the maxillary central incisors was used in the statistical analysis. Correlation coefficients between the photographic scores and the output from the software analyses were determined using for comparison with the metrics of ΔFch, Areach and ΔQch.
Frequency counts of subjects at each level of TF Index score
Photographic TF Score
City (frequency counts)
Intra class correlation coefficients for QLF metrics and mean metric values for each TF index score
QLF mETRIC (mean)
Kendall’s tau b
Description and frequency of subjects with additional compounding factors
Number of subjects
Missed demarcated opacity
Comparison of QLF metrics between cities
Sum of ranks
Mann Whitney U
Contingency table of subjects with and without fluorosis as determined by Δ Q (QLF) and photographic TF score
χ2 chi square
Fluorosis ΔQ ch
χ2 (1)= 31.735, P<0.0001
Fluorosis TF 0-2
χ2 (1)= 45.640, P<0.0001
Fluorosis TF 3-5
The purpose of this study was to use the QLF system within a standard epidemiological survey. The earlier work of Pretty et al. and use in larger populations identified strengths and weaknesses of fluorosis quantification by fluorescent imaging techniques. The encouraging results from the early work on intra class correlations between the QLF metrics and TF scores and the ability to detect differences in populations with different fluoride exposures gave justification for incorporating the system into an epidemiological survey. However, many of the issues raised by Pretty et al. remained unresolved.
The lack of an appropriate gold standard for comparison with the QLF metrics gave rise to statistical and interpretive problems as the data from the TF index is on an ordinal scale from 0 to 9 whereas the output from the QLF metrics generates continuous data. The consequence is there are no appropriate statistical methods to assess agreement. Hence, the use of correlations during analysis demonstrates the association between the outcomes, which should not be interpreted as agreement.
Choice of gold standard is not a unique issue to fluorosis quantification. QLF and other fluorescent imaging techniques have been used to detect caries with similar issues regarding agreement between outcome measures. In the case of caries detection, gold standards exist through histological examination using light microscopy and microradiography. These techniques have enabled the development of more robust assessment of agreement with QLF metrics relating to caries detection with cut off thresholds for the fluorescence devices. The validation of such devices for caries detection is an evolving subject influenced by the tooth surface under investigation and has been facilitated by the existence of more appropriate gold standards. The absence of an appropriate gold standard for fluorosis quantification resulted in a cut off threshold for ΔQch being determined from the ROC curve. This should not be interpreted as a transferable threshold for QLF analysis of other populations as it was not validated.
The decision to use the TF Index for fluorosis scoring was influenced by the index being based on the histological features associated with the presentation of the condition. However, fundamental differences exist between the aspects of the condition assessed by QLF and the TF Index. The former detects fluorosis over the whole tooth surface through fluorescence loss in image pixels whereas the latter assesses fluorosis not only from the clinical manifestations of histological changes but also from the patterns of presentation such as diffuse lines and confluent areas. Hence, the TF Index has no direct means of assessing the area of tooth surface involved. It is therefore interesting to find from the results of this study the QLF metric for area has the strongest correlation with the TF scores.
An inherent limitation of QLF is the inability to differentiate fluorescence loss as a result of fluorosis; other forms of developmental enamel defects and tooth surface phenomena such as enamel fractures and extrinsic stain. There is evidence to suggest that the use of computer software techniques may facilitate this process but this would involve more complicated image processing and tooth mapping prior to analysis.
The results of this study suggest that QLF has the ability to reliably quantify fluorosis in an epidemiological setting, albeit assisted by clinical diagnosis. In addition, QLF was able to discriminate between fluoridated and non-fluoridated populations. The intra class correlation coefficients are lower than those previously obtained[28, 30]. However, these associations are still significant and it should be stated that through each iterative stage of QLF evaluation the study populations have become larger, less selected, have demonstrated greater variety of fluorosis presentation and the potential for more confounding factors. Improved image mapping and software analysis may reduce these phenomena. Fluorescent imaging techniques such as QLF appear to have a high sensitivity but reduced specificity when employed in the detection and quantification of fluorosis impacting on the potential for these technologies to act as diagnostic tools for this condition. However, despite these limitations, QLF has the potential to monitor fluorosis longitudinally at both a population and individual level. Despite confounding factors the fluorescence imaging system may provide a useful objective, blinded system for the assessment of enamel fluorosis when used adjunctively with photographic scoring.
Area under the curve
Intra-class correlation coefficients
Light emitting diode
Quantitative light-induced fluorescence
Receiver operating characteristics
Statistical package for social sciences
Thylstrup & fejerskov index
- ΔFch :
Average fluorescence loss of areas considered fluorosis (convex hull technique)
- ΔQch :
Average fluorescence loss over entire tooth surface.
The authors wish to thank the pupils and staff at the schools who participated and Dianne Tabari, Deborah Howe and Valerie Dalton from Newcastle Primary Care Trust for their support and assistance during this project. The computer programmes used for image processing and remote scoring were created by Dr Andrew Taylor of the Dental Health Unit. IAP is funded by a Clinician Scientist Award from the National Institute for Health Research (UK). The Colgate Palmolive Dental Health Unit is funded by an unrestricted grant from Colgate Palmolive. This study was funded by the Clinician Science Award with support from The Colgate Palmolive Dental Health Unit.
- Whelton H, Crowley E, O’Mullane D, Donaldson M, Cronin M, Kelleher V: Dental caries and enamel fluorosis among the fluoridated population in the Republic of Ireland and non fluoridated population in Northern Ireland in 2002. Community Dent Health. 2006, 23 (1): 37-43.PubMedGoogle Scholar
- Whelton H, Crowley E, O’Mullane D, Donaldson M, Kelleher V, Cronin M: Dental caries and enamel fluorosis among the fluoridated and non-fluoridated populations in the Republic of Ireland in 2002. Community Dent Health. 2004, 21 (1): 37-44.PubMedGoogle Scholar
- Chankanka O, Levy SM, Warren JJ, Chalmers JM: A literature review of aesthetic perceptions of dental fluorosis and relationships with psychosocial aspects/oral health-related quality of life. Community Dent Oral Epidemiol. 2010, 38 (2): 97-109. 10.1111/j.1600-0528.2009.00507.x.View ArticlePubMedGoogle Scholar
- Clark DC: Trends in prevalence of dental fluorosis in North America. Community Dent Oral Epidemiol. 1994, 22 (3): 148-152. 10.1111/j.1600-0528.1994.tb01832.x.View ArticlePubMedGoogle Scholar
- NHS-CRD: A Systematic Review of Public Water Fluoridation. 2000, York: The University of York, NHS Centre for Reviews and Dissemination (Report 18)Google Scholar
- Tabari ED, Ellwood R, Rugg-Gunn AJ, Evans DJ, Davies RM: Dental fluorosis in permanent incisor teeth in relation to water fluoridation, social deprivation and toothpaste use in infancy. Br Dent J. 2000, 189 (4): 216-220.PubMedGoogle Scholar
- Ellwood RP, O’Mullane DM: Dental enamel opacities in three groups with varying levels of fluoride in their drinking water. Caries Res. 1995, 29 (2): 137-142. 10.1159/000262055.View ArticlePubMedGoogle Scholar
- Hamdan M, Rock WP: The prevalence of enamel mottling on incisor teeth in optimal fluoride and low fluoride communities in England. Community Dent Health. 1991, 8 (2): 111-119.PubMedGoogle Scholar
- Horowitz HS: The effectiveness of community water fluoridation in the United States. J Public Health Dent. 1996, 56 (5 Spec No): 253-258.View ArticlePubMedGoogle Scholar
- Dean HT, Arnold FA, Elvove E: Domestic water and dental caries. V. Additional studies of the relation of fluoride domestic waters to dental caries experience in 4,425 white children aged 12–14 years of 13 cities in 4 states. Public Health Rep. 1942, 57: 1155-1179. 10.2307/4584182.View ArticleGoogle Scholar
- Thylstrup A, Fejerskov O: Clinical appearance of dental fluorosis in permanent teeth in relation to histologic changes. Community Dent Oral Epidemiol. 1978, 6 (6): 315-328. 10.1111/j.1600-0528.1978.tb01173.x.View ArticlePubMedGoogle Scholar
- Angmar-Mansson B, de Josselin de Jong E, Sundstrom F, ten Bosch JJ: Strategies for improving the assessment of dental fluorosis: focus on optical techniques. Adv Dent Res. 1994, 8 (1): 75-79.PubMedGoogle Scholar
- Rozier RG: Epidemiologic indices for measuring the clinical manifestations of dental fluorosis: overview and critique. Adv Dent Res. 1994, 8 (1): 39-55.PubMedGoogle Scholar
- Dean HT: Classification of mottled enamel diagnosis. J Am Dent Assoc. 1934, 21: 1421-1426.Google Scholar
- Dean HT: Endemic Fluorosis and its relation to dental caries. Public Health report. 1938, 53: 1443-1452. 10.2307/4582632.View ArticleGoogle Scholar
- Dean HT: The investigation of physiological effects of the epidemiological method. Fluoride and Dental Health. Edited by: Moulton FR. 1942, Washington DC: American Association for the Advancement of Science, 23-3. AAAS Publ. No. 19Google Scholar
- Clarkson J: Review of terminology, classifications, and indices of developmental defects of enamel. Adv Dent Res. 1989, 3 (2): 104-109.PubMedGoogle Scholar
- Ellwood R, O’Mullane D, Clarkson J, Driscoll W: A comparison of information recorded using the Thylstrup Fejerskov index, Tooth Surface Index of Fluorosis and Developmental Defects of Enamel index. Int Dent J. 1994, 44 (6): 628-636.PubMedGoogle Scholar
- Tavener J, Davies RM, Ellwood RP: Agreement amongst examiners assessing dental fluorosis from digital photographs using the TF index. Community Dent Health. 2007, 24 (1): 21-25.PubMedGoogle Scholar
- Cochran JA, Ketley CE, Arnadottir IB, Fernandes B, Koletsi-Kounari H, Oila AM, van Loveren C, Whelton HP, O’Mullane DM: A comparison of the prevalence of fluorosis in 8-year-old children from seven European study sites using a standardized methodology. Community Dent Oral Epidemiol. 2004, 32 (Suppl 1): 28-33.View ArticlePubMedGoogle Scholar
- Cochran JA, Ketley CE, Duckworth RM, van Loveren C, Holbrook WP, Seppa L, Sanches L, Polychronopoulou A, O’Mullane DM: Development of a standardized method for comparing fluoride ingested from toothpaste by 1.5-3.5-year-old children in seven European countries. Part 2: Ingestion results. Community Dent Oral Epidemiol. 2004, 32 (Suppl 1): 47-53.View ArticlePubMedGoogle Scholar
- MRC: Medical Research Council Working Group Report. Water fluoridation and health. 2002, London: MRCGoogle Scholar
- Pretty IA: Caries detection and diagnosis: novel technologies. J Dent. 2006, 34 (10): 727-739. 10.1016/j.jdent.2006.06.001.View ArticlePubMedGoogle Scholar
- Angmar-Mansson B, ten Bosch JJ: Quantitative light-induced fluorescence (QLF): a method for assessment of incipient caries lesions. Dentomaxillofac Radiol. 2001, 30 (6): 298-307. 10.1038/sj.dmfr.4600644.View ArticlePubMedGoogle Scholar
- van der Veen MH, de Josselin de Jong E: Application of quantitative light-induced fluorescence for assessing early caries lesions. Monogr Oral Sci. 2000, 17: 144-162.View ArticlePubMedGoogle Scholar
- Pretty IA, Edgar WM, Higham SM: QLF quantification of plaque in a re-growth clinical trial. Caries Res. 2002, 36 (3): 214-215.Google Scholar
- Pretty IA, Edgar WM, Higham SM: The validation of quantitative light-induced fluorescence to quantify acid erosion of human enamel. Arch Oral Biol. 2004, 49 (4): 285-294. 10.1016/j.archoralbio.2003.11.008.View ArticlePubMedGoogle Scholar
- Pretty IA, Tavener JA, Browne D, Brettle DS, Whelton H, Ellwood RP: Quantification of dental fluorosis using fluorescence imaging. Caries Res. 2006, 40 (5): 426-434. 10.1159/000094289.View ArticlePubMedGoogle Scholar
- Taylor AM, Ellwood RP, Pretty IA, Mohan N: Quantitative stain detection in vivo using fluorescent imaging. J Dent. 2009, 37 (5): 397-405. 10.1016/j.jdent.2009.01.012.View ArticlePubMedGoogle Scholar
- McGrady MG, Ellwood RP, Srisilapanan P, Korwanich N, Taylor A, Goodwin M, Pretty IA: Dental fluorosis in populations from Chiang Mai, Thailand with different fluoride exposures. Paper 2: The ability of fluorescence imaging to detect differences in fluorosis prevalence and severity for different fluoride intakes from water. BMC Oral Health. 2012, 12 (1): 33-10.1186/1472-6831-12-33.View ArticlePubMedPubMed CentralGoogle Scholar
- Cochran JA, Ketley CE, Sanches L, Mamai-Homata E, Oila AM, Arnadottir IB, van Loveren C, Whelton HP, O’Mullane DM: A standardized photographic method for evaluating enamel opacities including fluorosis. Community Dent Oral Epidemiol. 2004, 32 (Suppl 1): 19-27.View ArticlePubMedGoogle Scholar
- ten Bosch JJ, Angmar-Mansson B: Characterization and validation of diagnostic methods. Monogr Oral Sci. 2000, 17: 174-189.View ArticlePubMedGoogle Scholar
- Kuhnisch J, Ifland S, Tranaeus S, Angmar-Mansson B, Hickel R, Stosser L, Heinrich-Weltzien R: Establishing quantitative light-induced fluorescence cut-offs for the detection of occlusal dentine lesions. Eur J Oral Sci. 2006, 114 (6): 483-488. 10.1111/j.1600-0722.2006.00404.x.View ArticlePubMedGoogle Scholar
- McGrady MG, Browne D, Ellwood RP, Pretty IA: Differentiation of demarcated enamel opacities from fluorosis using quantitative light-induced fluoroescence. Caries Res. 2008, 42: 213-Google Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1472-6831/12/47/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.