Clinical validation of robot simulation of toothbrushing - comparative plaque removal efficacy
© Lang et al.; licensee BioMed Central Ltd. 2014
Received: 16 January 2014
Accepted: 20 June 2014
Published: 4 July 2014
Clinical validation of laboratory toothbrushing tests has important advantages. It was, therefore, the aim to demonstrate correlation of tooth cleaning efficiency of a new robot brushing simulation technique with clinical plaque removal.
Clinical programme: 27 subjects received dental cleaning prior to 3-day-plaque-regrowth-interval. Plaque was stained, photographically documented and scored using planimetrical index. Subjects brushed teeth 33–47 with three techniques (horizontal, rotating, vertical), each for 20s buccally and for 20s orally in 3 consecutive intervals. The force was calibrated, the brushing technique was video supported. Two different brushes were randomly assigned to the subject. Robot programme: Clinical brushing programmes were transfered to a 6-axis-robot. Artificial teeth 33–47 were covered with plaque-simulating substrate. All brushing techniques were repeated 7 times, results were scored according to clinical planimetry. All data underwent statistical analysis by t-test, U-test and multivariate analysis.
The individual clinical cleaning patterns are well reproduced by the robot programmes. Differences in plaque removal are statistically significant for the two brushes, reproduced in clinical and robot data. Multivariate analysis confirms the higher cleaning efficiency for anterior teeth and for the buccal sites.
The robot tooth brushing simulation programme showed good correlation with clinically standardized tooth brushing.
This new robot brushing simulation programme can be used for rapid, reproducible laboratory testing of tooth cleaning.
KeywordsToothbrushing Plaque control Cleaning efficacy Robot simulation Randomized clinical trial Oral hygiene
Plaque removal by manual or powered toothbrushing is still the most effective preventive method to control gingivitis manifestation and caries lesion stagnation or progression . This cornerstone of oral hygiene is supported by lifelong local applications of different fluoride formulations and, when needed, of various antibacterial agents.
To motivate consumer’s oral hygiene behaviour and to enhance the patient’s compliance towards the recommended tooth cleaning efficacy, new toothbrush designs are permanently developed and tested. In contrast to the past, no “standard” toothbrush is dominating the market today. Preferences concerning handle configuration and brushhead design differ widely among preventive caregivers and consumers. Different age groups, patient’s disease profiles, patients in special needs etc. require individual toothbrush models and, consequently, individual brushing techniques in the frame of contemporary personalized preventive medicine. New toothbrush models for this various target groups should remove plaque equally efficient or better than their predecessors and, therefore, their plaque removal efficacy needs to be tested prior to manufacture.
The ultimate goal of such testing would be the outcome of clinical testing under field conditions by assessing full mouth plaque removal and gingivitis scores. This is, however, very time consuming, rather expensive and difficult to standardize for later comparative meta-analyses . Therefore, the assessment of in vitro tooth cleaning efficacy became a real alternative to clinical trials in testing many different designs and action modalities of manual and powered toothbrushes. Since 1972 several test environments were developed to test manual and powered toothbrushes prior to manufacture or clinical testing. Arnold and Trost were the first to introduce a simple brushing machine using horizontal movements on acrylic tooth models covered with a water-based dye . A more sophisticated equipment by Nygaard-Østby et al. was primarily developed to measure inter-proximal penetration of the toothbrush bristles by using a typewriter colour ribbon band to simulate inter-proximal space during horizontal or vertical brushing movements with brushing forces between 2.5 and 10.0 N . Rawls et al. proposed static and dynamic tests using recommended brushing techniques on blue ethyl cellulose coated typodont models at an angulation of 45°. Moistened toothbrushes were applied with weight controlled force between 1.0 and 10.0 N . The interproximal penetration of bristles was controlled and measured by a high-speed video camera and a colour removal index. The disadvantage of the experimental approach was the colour coating of plastic teeth not simulating the adherence of plaque biofilms on natural teeth. Therefore, Volpenhein et al. developed a plaque simulating red coating based on ethyl ester and copolymer. Manual toothbrushes were moved in two angulations (45° and 90º) and three directions (horizontal, vertical and rotating) over typodont models . Cleaning efficacy was assessed by the coating removal at a 10fold magnification of plastic teeth. According to the variability of the experimental protocol, this device was the first robot-like in vitro approach.
The first 6-axis robot was used to simulate the 3-dimensional brushing patterns of powered toothbrushes. The typodont models were stained with a water-based colour and brushed for 1 minute. A modified plaque index was used by two blinded examiners to score the results [6, 7]. The tooth cleaning efficacy assessment was later improved by a computerized vision system . Recently, the robot test of brush head wear was used to assess the area covered with plaque substitute by a 3 D laser system . Since a decade the brushing machine of Imfeld et al. is a well-established method of assessing in vitro tooth cleaning efficacy of manual and electric toothbrushes [10–12]. Test brushes were mounted on an automated brushing machine, which moved over a custom-made tooth model of a posterior or anterior segment. All black tooth surfaces were coated with white titanium oxide in ethanol to simulate complete plaque accumulation. Tooth surfaces reappearing black after brushing were regarded as cleaned, digitized and planimetrically analyzed.
The Navy-Plaque-Index was introduced by Elliot et al. to support the assessment of plaque in clinical studies by sectioning the buccal tooth surface into six areas (three gingival, and each one mesial, distal and incisal area) . Based on the need to test oral hygiene products the Navy-Plaque-Index was modified by Rustogi et al. by adding three additional zones above the gumline and below the equatorial line . Claydon and Addy modified this refined index by using planimetry for the examination process . After plaque revelation standardized clinical photographs were used to score all teeth buccally and orally with 576 planimetrical fields per subject. This resulted in better differentiation of plaque removal per tooth. Therefore, this planimetrical index is now well established, because of the reproducibility, blinded assessment and safe documentation.
In summary, there are sophisticated in vitro tooth cleaning methodologies, however, none of them have been clinically validated. And there are well established clinical index systems assessing reproducible plaque scores. It was, therefore, the aim of the present study to bridge the gap between the clinical performance and plaque removing outcome of tooth brushing and the in vitro simulation by the extraordinary programme flexibility of a six-axis robot.
The photographs of the clinical and robot programmes were categorized, blinded and scored by a calibrated examiner using the planimetrical index , with 9 fields at buccal sites and 9 fields at oral sites of all teeth (score 0 - no plaque; score 1 - plaque or plaque residues present).
The results of the planimetric scoring underwent statistical analysis by t-test, F-test and Wilcoxon signed-rank test. The outliers were detected prior to the analysis with the Grubbs’ test for outliers. To deal with the inhomogeneity of the sample values (n1 = 27, n2 = 14) a modification of the t-test (SATTERTHWAITE-WELCH-Test) was used to calculate the mean values of residual plaque between the subjects and the robot. The correlation of plaque removal efficacy of clinical and robot testing was detected with the Spearman rank correlation coefficient, and cleaning differences between the 3 brushing movements and the 2 toothbrushes were tested by multivariate analyses (see  for nonparametric statistics and  for t-tests and multivariate analysis).
Additional statistical tests were calculated excluding the planimetrical risk fields A, B and C next to the gumline to demonstrate the effect of gingival masks around typodonts. However, the number of significant differences in mean values of residual plaque between subjects and robot did not change, therefore, all 9 planimetrical fields per tooth site were included to calculate summary statistics of the plaque measurements for validation purposes only.
Comparison of mean values of residual plaque (not completely cleaned planimetrical fields, the maximum score of the planimetrical plaque Index is 9 (score 1 in each field A-I, the minimal score is =0 (no residual plaque in all 9 fields at the oral or buccal site)) in subjects and in robot testing
Both tested toothbrushes removed simulated plaque coatings in all 3 brushing movements with measurable statistical differences (brushing technique horizontally: lingual t = 4.55 (p < 0.01), buccal t = 4.04 (p < 0.01); brushing technique rotating: lingual t = 2.99 (0.01 < p < 0.05), buccal t = 5.90 (p < 0.01); brushing technique vertically: lingual t = −5.34 (p < 0.01), buccal t = −5.89 (p < 0.01). The average number of fields with remaining plaque ranged from 0.21 to 5.14 (horizontally), 0.64 to 4.50 (rotating) and from 2.00 to 4.33 (vertically). All robot data are represented in Table 1. The cleaning efficacy was again mainly tooth-specific and less site-specific except the two molars. Incisors, canines and premolars were much better cleaned compared to molars. Most buccal sites, except the second premolar, exhibited less simulated plaque compared to the lingual sites.
Robot vs. clinical plaque removal
with horizontal brushing movement (BMH), the use of toothbrush A (TBA), the lingual site (LING), incisors and canines (ABS1) and premolars (ABS2) as dummy-regressors.
Multivariate OLS-analysis of residual plaque in clinical and robot study – Estimated Coefficients
Plaque (Clinical data)
Plaque (Robot data)
6,74 (4 45***)
−15,57 (−8,38*** )
In particular, the estimated coefficients for the toothbrush are of great importance. In both cases (clinical and robot data), they significantly differ from zero with same sign. In the estimation of the clinical data, the use of toothbrush A leads to 6.74% more plaque than the use of toothbrush B (t = 4.45 (p < 0.01)). In the robot study it is 5.87%, also highly significant (t = 7.66 (p < 0.01)). The clinical results in favour of toothbrush B are reproduced in the robot study.
Laboratory testing of cleaning efficacy of different toothbrush designs is essential for the development of new prototypes as well as for the consumers interest in improving their individual oral hygiene. Therefore, any laboratory testing should be as close as possible to the real clinical conditions. In this sense, the robot testing approach has many advantages. This includes the programming (“teaching the robot”) and standardization of any brushing movement, the calibration of different brushing forces and brushing time. Planimetrical plaque index systems are applicable, and exploratory tests with 5 runs or statistical valid tests with 7 runs per test brush, per test movement etc. are possible. However, the clinical validation of the rather complex robot testing programme is a prerequisite for any clinical relevant conclusion concerning the plaque-biofilm controlling efficacy of toothbrushes. It was, therefore, the primary aim of the study to develop and validate a robot toothbrushing test in two steps.
The first step was the standardization of most recommended toothbrush movements of a common brushing force and brushing time and of a sensitive planimetrical plaque assessment index system applicable in the clinical and robot setting. Then, the translation of all this clinical parameters to the robot programme followed.
The second step was the comparison of clinical and robot plaque removal efficacy and the statistical approval whether the two data sets do correlate to declare the robot clinically validated.
The number of subjects was within the requirements for clinical toothbrush tests according to the ADA Acceptance Program Guidelines . The used brushing force of 3.5 N was slightly higher than the mean brushing force of uninstructed adults of 2.3 N (+/− 0.7) but within the range of an acceptable force of 4.0 N [21, 22]. The brushing time of 20 s buccally and 20 s orally per two sextants would sum-up in a total brushing time of 120 s for the complete dentition. This is slightly above a mean brushing time of uninstructed subjects of 96 s +/− 36 s .
All in vitro tests developed so far are capable to measure special aspects of plaque removal like inter-proximal penetration, brushing forces, brushing time and brushing technique, but lack the simulation of oral biofilms on tooth surfaces [2, 3, 7, 10]. None of the robot testing methodologies developed so far could satisfactorily reproduce the complex human tooth cleaning behaviour of rather sticky plaque biofilms.
According to the most comprehensive study of Claydon et al. of comparative professional plaque removal (using 8 branded toothbrushes) there was a clear tooth-specific and site-specific removal efficacy, but no significant differences between upper and lower arches . Consequently, in the present study two sextants of lower teeth including incisors, canines, premolars and molars were postulated to be representative for the whole dentition. The left lower canine and the right lower wisdom tooth were present but not scored to maintain the interproximal planimetrical fields. Claydon et al. did also show that there were no significant differences between the 8 toothbrushes. It supports also “the conclusion that the user is by far the most significant variable” .
The tooth-specificity and site-specificity of plaque accumulation and removal is caused by the size and morphology of teeth and by anterior vs. posterior teeth. Individual patterns may also be influenced by eugnathic vs. dysgnathic dentitions. The mean values of plaque removal tooth by tooth and site by site in the clinical study arm corresponded significantly with the mean values of the robot arm. However, the cleaning efficacy is significantly higher after robot brushing compared to clinical brushing. Therefore, the robot programme avoids any individual adaptation of brushing movements and brushing force different from the set standard. To test the brushing efficacy as such, the robot and the standardized programme seems to be superior to clinical testing as long as the toothbrush head design, the number and size of filaments, their stiffness and direction is concerned.
Abrasion, attrition and erosion, the tooth wear, is physically different from gentle brushing actions. So, a toothbrush is aimed at brushing (not abrading, not eroding) tooth biofilms partly away to control plaque accumulation. Rosema et al. have recently documented that “dentifrice did not show an added effect on instant plaque removal efficacy” using new and used manual toothbrushes . These results and other clinical studies  support the approach in the present study not to use dentifrice while searching for plaque or simulated plaque removal. The robot toothbrushing technology can be employed with gingival masks simulating the permanent and mixed dentitions and, by adapting the planimetrical index, without masks simulating gingival recession . Recently, the robot testing of children’s toothbrushes in the deciduous dentition has been clinically validated .
Due to short life cycles of new toothbrush models there is a need for quick and reliable testing of different prototypes and new models. One requirement before manufacturing a new toothbrush model is, that it should be at least as effective in plaque removal as its predecessor. Clinical tests normally cannot discriminate small differences between toothbrush models and therefore are not ideal in the early stage of development. The main advantage of the proposed robot test is the reproducibility of the test conditions and the precise data recording. Because of the clinical validation the robot methodology can predict clinical toothbrushing efficacy.
The individual data sets concerning the risk of residual plaque at teeth at risk, sites at risk and planimetrical fields at risk can be accumulated in databanks where they are comparable because of the standardization for many different forms, toothbrush prototypes and models on the market.
Planimetrical plaque scoring at all teeth of a representative artificial permanent dentition has been clinically validated. Therefore, the new robot test of toothbrushing can be recommended for the reproducible evaluation of plaque control and cleaning efficacy of different toothbrush designs and brushing actions.
- Deacon SA, Glenny AM, Deery C, Robinson PG, Heanue M, Walmsley AD, Shaw WC: Different powered toothbrushes for plaque control and gingival health. Cochrane Database of Systematic Reviews. 2010, Issue 12. Art. No.: CD004971. doi:10.1002/14651858.CD004971.pub2Google Scholar
- Arnold M, Trost G: Über die Abhängigkeit des Putzeffektes von verschieden Formen des Bürstenkopfes. Dtsch Stomatol. 1972, 22: 46-53.PubMedGoogle Scholar
- Nygaard-Østby P, Edwardsen S, Spydevold B: Access to interproximal tooth surfaces by different bristle designs and stiffness of toothbrushes. Scan J Dent Res. 1979, 87: 424-430.Google Scholar
- Rawls HR, Smith NK, Lentz DL, Cobb GW: Interproximal penetration of commercial toothbrushes as determined by static and dynamic tests using recommened brushing techniques. J Clin Dent. 1993, 4: 88-95.Google Scholar
- Volpenhein DW, Walsh MD, Dellermann PA, Burkett TA: A new method for in vitro evaluation of the interproximal penetration of manual toothbrushes. J Clin Dent. 1994, 5: 27-33.PubMedGoogle Scholar
- Ernst CP, Willershausen B, Driese G, Warren PR, Hilfinger P: A robot system for evaluating plaque removal efficiency of toothbrushes in vitro. Quintessence Int. 1997, 28: 441-445.PubMedGoogle Scholar
- Driesen GM, Warren PR, Hilfinger P, Ernst CP, Willershausen B: The development of the Braun Oral-B Ultra Plaque Remover: an in vitro robot study. Am J Dent. 1996, 9: 13-17.Google Scholar
- Driesen GM, Warren PR, Bielfeldt U, Helbig G: A laboratory comparison of the efficacy of battery-operated, non-rechargeable power toothbrushes. Am J Dent. 2001, 14 (Sec No): 5B-8B.PubMedGoogle Scholar
- Kaiser E, Thurnay S, Markgraf D, Pack S, Grender J, Hengenhold D, Warren PR, Wehrbein H, Erbe C: Brush head wear, subject-perceived and laboratory cleaning performance of two oscillating-rotating electric toothbrush heads over 3 months. Am J Dent. 2012, 25: 84-90.PubMedGoogle Scholar
- Imfeld T, Sener B, Simonovic I: In-Vitro-Untersuchung der mechanischen Wirkung von handelsüblichen Handzahnbürsten. Acta Med Dent Helv. 2000, 5: 37-47.Google Scholar
- Schaetzle M, Imfeld T, Sener B, Schmidlin PR: In vitro tooth cleaning efficacy of manual toothbrushes around brackets. Eur J Orthod. 2009, 31: 103-107.View ArticleGoogle Scholar
- Schaetzle M, Sener B, Schmidlin PR, Imfeld T, Attin T: In vitro tooth cleaning efficacy of electric toothbrushes around brackets. Eur J Orthod. 2010, 32: 481-489.View ArticleGoogle Scholar
- Elliot JR, Bowers GM, Clemmer BA, Rovelstad GH: Evaluation of an oral physiotherapy center in the reduction of bacterial plaque and periodontal disease. J Periodontol. 1972, 43: 221-224.View ArticleGoogle Scholar
- Rustogi KN, Curtis JP, Volpe AR, Kemp JH, McCool JJ, Korn LR: Refinement of the modified navy plaque index to increase plaque scoring efficiency in gumline and interproximal tooth areas. J Clin Dent. 1992, 3: 9-12.Google Scholar
- Claydon N, Addy M: The use of planimetry to record and score the modified Navy Index and other area-based plaque indices. A comparative toothbrush study. J Clin Periodontol. 1995, 22: 670-673.View ArticlePubMedGoogle Scholar
- Stagge P, Krollmann FH, Veith F: Report A 07 1086. 2007, Germany: MPA DarmstadtGoogle Scholar
- Sidney S, Castellan NJ: Nonparametric Statistics for the Behavioral Sciences. 1988, New York: McGraw Hill, SecondGoogle Scholar
- Gujarati DN: Basis Econometrics. 2003, New York: McGraw Hill, FourthGoogle Scholar
- Greene WH: Econometric Analysis. 1990, New York: Macmillan, SecondGoogle Scholar
- ADA American Dental Association: Acceptance Program Guidelines Toothbrushes. 2012, Council on Scientific Affairs, URL http://www.ada.org/sections/scienceAndResearch/pdfs/guide_toothbrushes.pdf [accessed May 15, 2014]Google Scholar
- Ganss C, Hardt M, Blazek D, Klimek J, Schlueter N: Effects of toothbrushing force on the mineral content and demineralized organic matrix of eroded dentine. Eur J Oral Sci. 2009, 117: 255-260.View ArticlePubMedGoogle Scholar
- Ganss C, Schlueter N, Preiss S, Klimek J: Tooth brushing habits in uninstructed adults - frequency, technique, duration and force. Clin Oral Invest. 2009, 13: 203-218.View ArticleGoogle Scholar
- Claydon N, Addy M, Scratcher C, Ley F, Newcombe R: Comparative professional plaque removal study using 8 branded toothbrushes. J Clin Periodontol. 2002, 29: 310-316.View ArticlePubMedGoogle Scholar
- Rosema M, Hennequin-Hoenderdos NL, Versteeg PA, Van der Weijden GA: Plaque removing efficacy of new and used manual toothbrushes - a professional brushing study. Int J Dent Hygiene. 2013, 11: 237-243.View ArticleGoogle Scholar
- Gaengler P, Lang T, Jennes B: Computer-assisted planimetrical plaque assessment of robot tested toothbrushing. J Dent Res. 2013, 92 (Spec. Issue): Abstract No. 3326Google Scholar
- Lang T, Jennes B, Gaengler P: Clinical validation of robot tested toothbrushing of deciduous dentition. J Dent Res. 2014, 93 (Spec. Issue): Abstract No. 494Google Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1472-6831/14/82/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.