Less extrusion debris during the retreatment of curved canals using twisted files with higher rotational speeds: an ex vivo study
© The Author(s). 2017
Received: 27 June 2016
Accepted: 10 January 2017
Published: 16 January 2017
Debris extrusion from the apical foramen can be problematic in severely curved canals. This study aimed to assess the use of Twisted Files (TF) at different rotational speeds during retreatment, compared with manual technique using Triple-Flex Files (TFF).
Forty-eight mesiobuccal root canals were randomized to four groups (n = 12 per group). In groups A, B, and C, gutta-percha was removed using TF at 500, 1000, and 1500 rpm, respectively, while it was removed using TFF in group D. Apical debris was collected in a pre-weighed centrifuge tube. The weight of dry debris was assessed by comparing the pre- and post-instrumentation weight for each group.
Instrumentation in group D (0.69 ± 0.04 mg) extruded significantly more debris than any of the TF groups (A: 0.54 ± 0.05 mg; B: 0.48 ± 0.04 mg; C: 0.42 ± 0.03 mg; all P < 0.001). In addition, increasing the rotational speed of TF decreased the amount of extruded debris (A vs. B: P = 0.006; B vs. C: P < 0.001; A vs. C: P < 0.001).
TF at 1500 rpm produced less apical extrusion debris than other TF operating speeds and TFF.
KeywordsApical debris Root canal therapy Rotational speed Tooth apex Triple-Flex Files Twisted Files
The main goal of endodontic retreatment is to remove infected dentin and root-filling material throughout the canal length to re-establish decontamination of the root canal system and to preserve the tooth [1, 2]. Debris generated during retreatment may cause undesired consequences, such as inflammation and postoperative pain [1, 3]. Many techniques have been used to remove gutta-percha in root-filled teeth, such as hand files, heat, solvents, and rotary instruments . Unfortunately there is considerable evidence that all of these techniques apically extrude some of the debris, but the amount is likely to vary according to the technique used [4–6]. Motor-driven rotary devices are associated with less dentinal debris extrusion because of their unique file design that also works as an Archimedic screw , but the complexity of root canal anatomy makes retreatment difficult and highlights the need for investigation of these issues. Indeed, when using rotary instruments, the high degree of root canal curvature may lead to iatrogenic damage, and this risk in increased with higher rotational speed. A proper evaluation of the different tools available would then allow appropriate and safe retreatment techniques to be selected with the lowest incidence of extrusion possible.
Twisted Files (TF; SybronEndo, Orange, CA, USA) are rotary nickel titanium endodontic files that were introduced in 2008. They can be twisted and converted back into an austenite structure by heating and cooling . The manufacturer claims that TF are superior to traditional file systems due to their cyclic fatigue resistance, flexibility, and better cutting efficiency . Compared with other instruments, TF can work safely at higher rotational speeds, but few studies have investigated the behavior of TF during retreatment . Previous retreatment studies have been performed on relatively straight root canals [11–13], which may not accurately represent the challenge of retreating curved canals. Furthermore, almost all rotary instruments were studied at similar speeds. Therefore, the influence of rotational speed on effective retreatment has not been assessed.
The purpose of the present study was to quantitatively evaluate the amount of debris extruded apically after endodontic retreatment with TF at different operating speeds, compared with traditional hand techniques using Triple-Flex Files in severely curved root canals.
A total of 48 mandibular first molars were collected from patients with periodontal disease undergoing tooth extraction. To be included in the study, tooth had to be without a history or disease or treatment. This study was approved by the ethics committee of the Affiliated Hospital of Qingdao University.
Radiographs of each tooth were obtained, digitized, and stored electronically using the Digora Optime digital imaging system (Soredex, Tuusula, Finland). Angles and radius of curvature were measured using the proprietary software Digora for Windows (DfW version 2.7, Soredex, Tuusula, Finland). Root canal curvature was determined according to the Schneider’s method . The radius of each sample was measured according to Schäfer et al. .
The mesiobuccal root canals were used when encountering the following criteria: 1) curvature ranging between 25° and 35°; 2) a radius below 10 mm in a buccal direction; and 3) not more than 5° in a mesial to distal direction. The mesiobuccal root canals were examined under an operating microscope (Carl Zeiss, Jena, Germany) to verify the presence of a single apical foramen, with fully formed apices. The roots had no previous root canal treatment. The access cavity was prepared using high-speed diamond burs (Dentsply Maillefer, Ballaigues, Switzerland) with water cooling. After accessing the cavity, the canal patency was established with a size 10 K-file (Dentsply Maillefer, Ballaigues, Switzerland). Only canals with an initial apical size equivalent to a size 10 K-file were selected.
According to the above criteria, 48 molars with severely curved mesiobuccal root canals were collected and thoroughly cleaned by removing the calculus and the soft deposits using curettes. The teeth were stored in a saline solution before use.
Root canal preparation and obturation
Apical patencies were determined with a size 10 K-file. The K-file was introduced carefully into the canal until it was visible at the apical foramen, and the working length (WL) was established as 1 mm short of this length. The crowns were flattened, so that a final dimension of 13-mm working length was achieved for each root.
The coronal portion was initially flared with Gates Glidden drills (Dentsply Maillefer, Ballaigues, Switzerland) sizes 1–3 and was instrumented using ProTaper Universal Rotary Files (Dentsply Maillefer, Ballaigues, Switzerland) up to size F1, according to the manufacturer’s instructions. Each instrument was used for the preparation of only one tooth. On withdrawal of each instrument, the canals were flushed with 5 mL of distilled water. ProTaper Universal Gutta Percha Point size F1 (Dentsply Maillefer, Ballaigues, Switzerland) was selected as the master gutta-percha cone fitted with tug-back at the WL. The root canals were dried with sterile paper points and obturated with the master cone and sealer using a continuous wave of condensation technique (System B, Elements Obturation Unit, Sybron Endo, Orange, CA, USA). The total length of the root canal filling did not exceed the level of the staging platform from the apex to the coronal aspect, and also assessed by anteroposterior and lateral view X-ray photography, in an attempt to control the amount of root filling in each tooth, so that the volume of the filling material was approximately equal in all root canals.
The teeth were radiographed (Soredex Minray, Soredex, Milwaukee, WI, USA) at different angles to confirm the quality of the filling procedure. The access cavities were temporarily filled (Cavition-GC, Tokyo, Japan). The obturated roots were stored in 100% humidity at 37°C for two weeks to allow complete setting of the sealer.
All teeth were coded and randomized to four groups (12 teeth/group). Gutta-percha in the coronal portion was removed using size 1–3 Gates Glidden drills. The rotary instrument was driven by an endodontic motor (NSK, Endo-Mate DT, Tokyo, Japan) with the crown-down technique at the recommended speed and torque according to the manufacturer’s instructions. The final apical file was one size larger than that used in the first preparation. The four groups were then retreated according to their group allocation. No apical counter pressure was applied.
Group A: Specimens were retreated using TF at 500 rpm without torque-control, according to the manufacturer’s recommendations, and using a 4:1 speed reduction and an electric motor at 2000 rpm. The instrumentation sequence was .01/#25, .06/#25, and .08/#25 at the WL. The shaping procedure began with TF size .04/#25. TF size 25/08 was inserted and used as the final apical file.
Group B: TF at 1000 rpm. The method for gutta-percha removal was the same as in group A except the motor was used with 4:1 speed reduction and was powered by an electric motor at 4000 rpm, resulting in instrument rotation speed of 1000 rpm.
Group C: TF at 1500 rpm. The method for gutta-percha removal was the same as in group A except the motor was used with 4:1 speed reduction and was powered by an electric motor at 6000 rpm, resulting in instrument rotation of 1500 rpm.
Group D: manual technique. The canals were reinstrumented sequentially using triangular Triple-Flex Files (SybronEndo, Orange, CA, USA; made of stainless steel designed for flexibility and integrity) in sizes 20 to 25, to a master apical file size of 25. The files were pre-curved according to root canal direction and gutta-percha was removed from the canal using the balance force technique and step-back in 0.625-mm increments (determined using a caliper) to a file size 55 so that the taper was determined to be 8%. Canal patency was maintained with a size 25 K-file during reinstrumentation. Consequently, the final canal diameter and taper was comparable to the samples in the other groups.
During all retreatment procedures, the flutes of the files were cleaned completely after each use. Canals were flushed with 20 mL of distilled water. All instruments were used for only one specimen and then discarded. Any deformed instruments were also discarded. Retreatment was deemed complete when the WL was reached and no more gutta-percha was visible on the last instrument used in every group, based on the step-back approach .
After retreatment, any debris (dentin and filling material) visually adherent to the external surface of the apex was scraped and collected into the tube containing the sample. The root apex was flushed with 0.5 mL of distilled water to wash any remaining debris (dentin and filling material) into its tube. No solvent was used. Each root canal was prepared, filled, and retreated by the same operator to reduce interoperator variability.
After the retreatment procedure, the centrifuge tubes were removed from the vials by handling with clean cotton forceps at all times. They were stored in an incubator at 37°C for 4 weeks to evaporate moisture before finally being weighed three times on a microbalance and the average value was used for analysis. The weight of the extruded debris was determined by subtracting the weight of the preweighed empty tube from the final weight. Evaluation was carried out by a second examiner who was blinded to group assignment.
Statistical analysis was performed using SPSS 19.0 (IBM, Armonk, NY, USA). In the preliminary analysis, the raw data followed a normal distribution (Shapiro-Wilk test); statistical analysis was thus performed using parametric one-way analysis of variance. Post-hoc pairwise comparisons were performed using the Tukey’s multiple comparison test. Statistical significance was established at 0.05.
Characteristics of the teeth
Angle and radius of canal curvature of the 48 teeth
Mean angle (degree)
Mean radius (mm)
A(n = 12)
B(n = 12)
C(n = 12)
D(n = 12)
Amount of debris
Amount of Apically Extruded Debris
Mean Weight (mg)
The aim of this study was to evaluate the use of TF for endodontic retreatment in terms of prevention of extrusion debris and in cases of severely curved mesiobuccal root canals. To do so, three different speeds of rotation were compared when using the TF and manual treatment of curved root canals. The results showed that each of the methods used produced extrusion debris, but that TF at 1500 rpm produced the smallest amount.
Reducing extrusion debris is important for effective endodontic retreatment as it reduces the likelihood of inflammation and postoperative pain. These results are consistent with previous retreatment studies that used various nickel titanium rotary instruments compared with conventional instruments, and demonstrated that rotary instruments produced less debris . This is due, at least in part, to the up-and-down motion of the manual files, which is more likely to push debris toward the apical end .
Previous studies on retreatments have been performed on relatively straight root canals [11–13], because their low curvatures eliminated complications likely to arise in the instrumentation of severely curved root canals. However, it can be argued that the filling materials in straight root canals are more easily removed so that canal repreparation tends to extrude less debris, and this may not accurately represent the challenge of retreating curved canals. On the contrary, teeth with high degrees of curvature may give rise to different results . Therefore, studying these devices in straight root canals only does not fully reflect the differences among various technologies. It has been shown that in mandibular first molars’ mesiobuccal root canals the mean curvature is 25° and the mean radius is 10.6 mm . Thus, the present study investigated mesiobuccal root canals with severe curvature angles of 25–35° and a short radius of <10 mm in an attempt to simulate these challenging clinical conditions. To the best of the authors’ knowledge, there have been no peer-reviewed studies in which molars with severely curved roots were used to assess the amount of dentin debris extruded during retreatment.
In previous studies, the continuous wave of condensation technique  has been rarely used in curved canals. We used this method before retreatment to ensure filling and to avoid filling defects, to make that the results of the investigation are more reliable. The technique has become increasingly popular because it uses heat to produce a homogenous obturation that adapts well to the canal walls and replicates the prepared root-canal space [20, 21].
Unlike previous studies, a major factor in the present study was the use of different rotational speeds. High rotational speed not only increases efficiency but also probably produces more heat . This may be because rotary reinstrumentation generates frictional heat during its contact with gutta-percha and dentin. In addition, rotational speed is associated with heat generation during the formation of martensite . The resulting heat could plasticize the gutta-percha, making it into a single block, thus facilitating its removal and minimizing extruded debris . In addition, plasticized gutta-percha cannot pass through the apical foramen , which could have contributed to the differences observed at high rotational speed. Furthermore, the Archimedes screw effect due to higher rotational speed should be taken into account . On the other hand, the exact same methods for coronal introduction of gutta-percha was used among all groups, and eventual differences in pressure should be counterbalanced by the standard method and randomization of the samples. Nevertheless, the plasticization of gutta-percha due to heat during retreatment could be associated with thermal damage to the teeth , but heat damage was not examined in the present study. Additional studies are necessary to determine if heat due to rotational speed during retreatment is associated with significant heat damage.
Although the current ground-fluted rotary instruments that are commonly used have excellent performance, there are inevitably many drawbacks due to their design. They have difficulties negotiating a severe curvature, which may increase the risk of iatrogenic damage such as bleeding and perforation of the lateral wall of the root canal. Furthermore, they must be operated in strict accordance with the speed and torque instructions, otherwise there is a risk of unexpected instrument separation, especially in curved root canals . Since most devices have similar low rotational speeds, and the difference is very small, comparisons between device speeds are almost meaningless. Rotational devices that can be used effectively in curved root canals and at higher speeds could improve retreatment.
TF, which are manufactured by twisting instead of grinding the nickel titanium alloy, have been shown to have excellent performance . TF with R-phase technology gives a much higher level of flexibility and can perform side-cutting with great efficiency, while still successfully negotiating a complex curvature. Above all, TF can work at speeds up to 1500 rpm, underlining that using higher rotational speed will lead to less debris. The manufacturer suggests that using a speed of 500 rpm is likely to reduce waste and increase the service life of the instrument, but our experience of using it between 900 and 1500 rpm is also positive and more efficient. We therefore tested TF at three different speeds 500, 1000, and 1500 rpm to assess whether speed would influence the amount of apical debris. The results showed that TF at the highest rotational speed produced the least debris. The results of the experiment were consistent with our inference of the benefits of increased speed and temperature. Nevertheless, other instruments use different rotational speeds and additional studies are necessary to assess these instruments.
This study has some limitations. The clinical relevance of this ex vivo study should be interpreted with caution. In vivo, the apex would be surrounded by periradicular tissues that may serve as a natural barrier preventing debris extrusion. In an early (1977) in vivo study, Salzgeber and Brilliant  showed that vital tissues helped control the apical and lateral penetration of an irrigating solution. In addition to the apical control of extruded debris, other factors involved in retreatment require further evaluation. For instance, the heat produced by high speed rotation may be transmitted to the outer root. It has been shown that a 10°C rise may cause irreversible thermal damage to the supporting periodontal structures . Whether the heat produced in this investigation could cause damage to the surrounding tissue needs further evaluation. Finally, apart the power tool nature of TF, there was intrinsic differences between the TF and the manual instrument, such as the cross section (round for TF and triangular for manual files), alloy, and the number of files used. Nevertheless, groups A, B, and C used the same instrument. Additional research will be necessary to evaluate the cleaning efficiency of the TF at higher speeds and to determine the optimal rotational speeds (and torque values) for the removal of gutta-percha.
These results provide important information on preventing apical debris and should help improve retreatment outcomes. TF at high rotational speed was associated with less debris extrusion than at lower speeds, and in comparison with manual endodontic retreatment in curved root canals. Speed may be a significant factor for the retreatment results and may improve retreatment outcomes. Further research is needed to fully evaluate the effect of high rotational speed on endodontic retreatment.
Availability of data and materials
The de-identified data from this study can be requested from the corresponding author at: firstname.lastname@example.org. Individuals who are interested in accessing study data should submit a written request outlining their aims and analytical plans in obtaining and using the data with an address, contact and e-mail information for tracking purposes. The information provided to users should not be used for commercial purposes, and should not be distributed to third parties.
MDL participated in literature search, study design, data collection, data analysis, and wrote the manuscript. SJX conceived and coordinated the study, analyzed the data, and provided a critical revision. FT and YL carried out the data collection and statistical analysis. All authors reviewed the results and approved the final version of the manuscript.
The authors declare that they have no competing interests.
Consent for publication
Ethics approval and consent to participate
This study was approved by the ethics committee of the Affiliated Hospital of Qingdao University. The need for informed consent was waived by the committee.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Torabinejad M, Corr R, Handysides R, Shabahang S. Outcomes of nonsurgical retreatment and endodontic surgery: a systematic review. J Endod. 2009;35:930–7.View ArticlePubMedGoogle Scholar
- Kim S, Kratchman S. Modern endodontic surgery concepts and practice: a review. J Endod. 2006;32:601–23.View ArticlePubMedGoogle Scholar
- Siqueira Jr JF. Microbial causes of endodontic flare-ups. Int Endod J. 2003;36:453–63.View ArticlePubMedGoogle Scholar
- Lumley PJ, Adams N, Tomson P. Root canal retreatment. Dent Update. 2006;33:210–5.Google Scholar
- Huang X, Ling J, Wei X, Gu L. Quantitative evaluation of debris extruded apically by using ProTaper Universal Tulsa rotary system in endodontic retreatment. J Endod. 2007;33:1102–5.View ArticlePubMedGoogle Scholar
- Saad AY, Al-Hadlaq SM, Al-Katheeri NH. Efficacy of two rotary NiTi instruments in the removal of Gutta-Percha during root canal retreatment. J Endod. 2007;33:38–41.View ArticlePubMedGoogle Scholar
- Ferraz CC, Gomes NV, Gomes BP, Zaia AA, Teixeira FB, Souza-Filho FJ. Apical extrusion of debris and irrigants using two hand and three engine-driven instrumentation techniques. Int Endod J. 2001;34:354–8.View ArticlePubMedGoogle Scholar
- Gambarini G, Grande NM, Plotino G, Somma F, Garala M, De Luca M, et al. Fatigue resistance of engine-driven rotary nickel-titanium instruments produced by new manufacturing methods. J Endod. 2008;34:1003–5.View ArticlePubMedGoogle Scholar
- Kim HC, Yum J, Hur B, Cheung GS. Cyclic fatigue and fracture characteristics of ground and twisted nickel-titanium rotary files. J Endod. 2010;36:147–52.View ArticlePubMedGoogle Scholar
- Marfisi K, Mercade M, Plotino G, Duran-Sindreu F, Bueno R, Roig M. Efficacy of three different rotary files to remove gutta-percha and Resilon from root canals. Int Endod J. 2010;43:1022–8.View ArticlePubMedGoogle Scholar
- Burklein S, Schafer E. Apically extruded debris with reciprocating single-file and full-sequence rotary instrumentation systems. J Endod. 2012;38:850–2.View ArticlePubMedGoogle Scholar
- Kustarci A, Akpinar KE, Er K. Apical extrusion of intracanal debris and irrigant following use of various instrumentation techniques. Oral Surg Oral Med Oral Pathol Oral Radiol Endod. 2008;105:257–62.View ArticlePubMedGoogle Scholar
- Surakanti JR, Venkata RC, Vemisetty HK, Dandolu RK, Jaya NK, Thota S. Comparative evaluation of apically extruded debris during root canal preparation using ProTaper, Hyflex and Waveone rotary systems. J Conserv Dent. 2014;17:129–32.View ArticlePubMedPubMed CentralGoogle Scholar
- Schneider SW. A comparison of canal preparations in straight and curved root canals. Oral Surg Oral Med Oral Pathol. 1971;32:271–5.View ArticlePubMedGoogle Scholar
- Schafer E, Diez C, Hoppe W, Tepel J. Roentgenographic investigation of frequency and degree of canal curvatures in human permanent teeth. J Endod. 2002;28:211–6.View ArticlePubMedGoogle Scholar
- Myers GL, Montgomery S. A comparison of weights of debris extruded apically by conventional filing and Canal Master techniques. J Endod. 1991;17:275–9.View ArticlePubMedGoogle Scholar
- Endodontics CP. Part 7. Preparing the root canal. Br Dent J. 2004;197:603–13.View ArticleGoogle Scholar
- Tanalp J, Kaptan F, Sert S, Kayahan B, Bayirl G. Quantitative evaluation of the amount of apically extruded debris using 3 different rotary instrumentation systems. Oral Surg Oral Med Oral Pathol Oral Radiol Endod. 2006;101:250–7.View ArticlePubMedGoogle Scholar
- Buchanan LS. The continuous wave of condensation technique: a convergence of conceptual and procedural advances in obturation. Dent Today. 1994;13:80. 2, 4-5.PubMedGoogle Scholar
- Wu MK, Kast'akova A, Wesselink PR. Quality of cold and warm gutta-percha fillings in oval canals in mandibular premolars. Int Endod J. 2001;34:485–91.View ArticlePubMedGoogle Scholar
- Lea CS, Apicella MJ, Mines P, Yancich PP, Parker MH. Comparison of the obturation density of cold lateral compaction versus warm vertical compaction using the continuous wave of condensation technique. J Endod. 2005;31:37–9.View ArticlePubMedGoogle Scholar
- Bramante CM, Fidelis NS, Assumpcao TS, Bernardineli N, Garcia RB, Bramante AS, et al. Heat release, time required, and cleaning ability of MTwo R and ProTaper universal retreatment systems in the removal of filling material. J Endod. 2010;36:1870–3.View ArticlePubMedGoogle Scholar
- Eggeler G, Hornbogen E, Yawny A, Heckmann A, Wagner M. Structural and functional fatigue of NiTi shape memory alloys. Mater Sci Eng A. 2004;378:24–33.View ArticleGoogle Scholar
- Betti LV, Bramante CM. Quantec SC rotary instruments versus hand files for gutta-percha removal in root canal retreatment. Int Endod J. 2001;34:514–9.View ArticlePubMedGoogle Scholar
- Vaudt J, Bitter K, Neumann K, Kielbassa AM. Ex vivo study on root canal instrumentation of two rotary nickel-titanium systems in comparison to stainless steel hand instruments. Int Endod J. 2009;42:22–33.View ArticlePubMedGoogle Scholar
- Chauhan A, Nilker V, Mandke LP. Effect of Temperature Rise on Periodontal Tissue During Endodontic Treatment: An In Vitro Study. Indian J Oral Health Res. 2015;1:66–71.View ArticleGoogle Scholar
- Zelada G, Varela P, Martin B, Bahillo JG, Magan F, Ahn S. The effect of rotational speed and the curvature of root canals on the breakage of rotary endodontic instruments. J Endod. 2002;28:540–2.View ArticlePubMedGoogle Scholar
- Salzgeber RM, Brilliant JD. An in vivo evaluation of the penetration of an irrigating solution in root canals. J Endod. 1977;3:394–8.View ArticlePubMedGoogle Scholar
- Eriksson AR, Albrektsson T. Temperature threshold levels for heat-induced bone tissue injury: a vital-microscopic study in the rabbit. J Prosthet Dent. 1983;50:101–7.View ArticlePubMedGoogle Scholar